2018-04-05 05:13:04 UTC
I have 2 10MHz channels, 1 and 2, software Acquire---HP 53131A/53132A/53181A---start measurement, but test ADEV is wrong. The curves of each measurement vary greatly.
time-nuts mailing list -- firstname.lastname@example.org
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
and follow the instructions there.