2018-05-11 17:12:58 UTC
Check TWO things.
1.) Has the RS-232 level converter (1488/1489, MAX232, etc.) failed?
2.) What MODE is your Z3816A, SCPI? The GPSCon software places Z3816A into SCPI.
This is discussed in K8CU’s notes, below:
K8CU Notes - HP Z3816A GPS Frequency Standard Receiver
I am getting some very peculiar behavior with the serial interface on
my GPS rcvr (Z3816A).
To date, the only thing that works is Sat Stat on a windows 2000 computer.
I am using a Null modem and set the monitor computer to 9600-8N1.
Other computers and/or utilities seem to give irregular behavior.
I have looked at the signals with a scope and they seem borderline OK.
The signals (Rx and Tx) swing a bit more than +/- 7 V. There is also some
digital noise riding on the lines.
I also notice that the -15 Vdc line on the internal power supply is not
working (it reads about + 1.2 V).
Someone on the web I found a posting claiming that the -15 Vdc line is not
needed for the Z3816A.
Any help would be appreciated.
1. Is 9600-8N1 correct?
2. Could the lack of the -15 Vdc line be the root cause?
3. Has anyone managed to use GpsCon with a Z3816A?
Thanks much for any input!
time-nuts mailing list -- email@example.com
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
and follow the instructions there.